The 1-year results of the Harmony transcatheter pulmonary valve to treat severe pulmonary regurgitation have shown a high rate of eliminating or reducing the degree of symptoms as well as freedom from endocarditis, sustained ventricular tachycardia, and the need for further interventions.
"Simply put, the good news is no endocarditis," said Daniel S. Levi, MD, in presenting results from three different studies with 108 patients who received three different iterations of the device at the Society for Cardiovascular Angiography & Interventions annual scientific sessions.
"Endocarditis has been an issue for us in the pulmonary position; we have yet to have an endocarditis in these patients in 1 year," he stressed.
The studies evaluated three different versions of the Harmony valve: TPV22 (42 patients), the first version with a 22-mm diameter; the Clinical TPV25 (17 patients), the first iteration of a 25 mm–wide device that has since been discontinued; and the modified TPV25 (45 patients), the second version of the 25-mm valve. The three studies are the early feasibility study of the TPV22, the continued-access study of the TPV22 and the mTPV25, and the pivotal study that included all three versions.
At baseline, 89% of patients had severe and 11% had moderate pulmonary regurgitation (PR). At 1 year, 92% had none or trace PR, 3% had mild PR, and 4% moderate disease.
Levi said the device "speaks for itself" in the results he presented. They include no deaths, no heart attacks, and no pulmonary thromboembolism. Other key outcomes include:
One major stent fracture in one of the early feasibility study patients at 1-month follow-up.
Four explants, with two in the discontinued cTPV25 and two with the TPV22 in the early-feasibility study.
Four reinterventions, two with the discontinued cTPV25 and two valve-in-valve procedures with the mTPV25 in the continued-access study, one with stent placement in the right ventricular outflow tract.
Levi and coinvestigators also performed a breakdown of 1-year outcomes – freedom from PR, stenosis, and interventions – by device: 95.1% for TPV22; 89.7% for mTPV25; and 73.3% for the discontinued cTPV25.
Although the valve is indicated for adolescents and adults, most of the patients in the three studies were adults, with an average weight of 165 pounds (75 kg) who have had PR for decades, said Levi, an interventional pediatric cardiologist at the University of California, Los Angeles. "With a device like this we are hopefully shifting to treating that a little bit earlier, but fortunately we don't usually need to treat it before puberty." The 25-mm TPV gives "a really nice landing zone" for future valve placement. "The goal is to keep patients out of the operating room for at least a few decades if not their whole lives," he said.
Levi said the Harmony investigators will follow outcomes with the 22- and modified 25-mm Harmony valves, both of which remain commercially available, out to 10 years.
The study represents the first collective cohort evaluating the Harmony device across the early feasibility, continued access and pivotal studies, said Brian Morray, MD. "It's important that people understand that evolution and how that impacts the way we look at outcomes, because when you aggregate the data, particularly for the TPV25, some of the procedural outcomes and the adverse events are no longer really reflective in the current time frame."
These Harmony results "represent another big step in the evolution of interventional cardiology and will be up there with development of the Melody valve and the utility and the use of the Sapien valve in the pulmonary position," said Morray, an associate professor of pediatrics at the University of Washington, Seattle, and an interventional cardiologist at Seattle Children's Hospital.
Levi disclosed he is a consultant to Medtronic and Edwards Lifesciences. Morray disclosed he is a clinical proctor for Abbott and a consultant to Medtronic, but not for the Harmony device.
This article originally appeared on MDedge.com, part of the Medscape Professional Network.
Lead image: Zerbor/Dreamstime
Medscape Medical News © 2022 WebMD, LLC
Cite this: Harmony Pulmonary Valve Update: Regurgitation Resolved 1 Year Out - Medscape - Jun 09, 2022.